Three-program transcranial impulse electrostimulator Transair-04
This Russian-made device is used in medical institutions of various types: general hospitals, clinics, medical centers of various profiles, etc. It is especially recommended for use in cases where there is a need to use a current strength of more than 3 mA. For example: with severe pain, treatment of alcoholism, drug addiction.
For healthy people
Maintaining a normal psycho-physiological state, the possibility of gaining psychological stability under any circumstances
Removal of pain of various origins and acceleration of healing. The result of anesthesia can be noticeable after the first procedure.
Treatment of depression
Effective application of the method in the treatment of various depressive conditions due to the production of stress-limiting hormones endorphins and serotonin
This model has 3 operating modes, an extended set of service functions, in particular, an auto-check mode, a session time indicator (with a countdown), frequency control. The device circuit includes an electronic patient protection system (automatic shutdown in case of malfunction).
The absence of research regimes makes this model easier to learn and operate, including lower polyclinics, feldsher-obstetric stations, etc.
The device is equipped with a CD with a session of psychomusic therapy.
Types of current: pulsed bipolar, pulsed monopolar and a combination of monopolar and direct currents in a combination of 1: 1
A feature of the electrical impact, carried out with the help of TRANSAIR-04 devices, is stimulation with special electric rectangular current pulses of a fixed frequency and duration. The position of the electrodes on the head is also strictly fixed – the negative electrode is always located in the forehead area, the positive one is also behind the ears. Thus, the patient is not required to select the exposure parameters except for the value of the therapeutic current) and the position of the electrodes, which greatly simplifies the treatment process and makes it safe.
This type of transcranial electrical stimulation has received a special name in the scientific and medical literature – TES therapy.. It is proved that the current from the device penetrates through the skin and soft tissues of the head, the skull and ultimately acts on the antinociceptive structures of the brain. This ensures the activation of the protective mechanisms of the brain.
It has been shown that already 10-15 minutes after the start of stimulation, there is an increase in the release of opioid peptides (ß-endorphin) and an increase in their concentration in the brain, cerebrospinal fluid and blood several times. , serotonergic and cholinergic neurotransmitter mechanisms are also involved.
- Form of stimulating current:
• rectangular bipolar pulses;
• rectangular monopolar pulses;
• direct current;
• combination of rectangular monopolar pulses with direct current
- Output signal adjustment range:
• rectangular bipolar pulses up to 5 mA;
• rectangular monopolar pulses up to 5 mA;
• constant current up to 5 mA;
• combination of rectangular monopolar pulses with direct current up to 10 mA
Frequency deviation range with frequency modulation of 74-80 Hz;
The duration of the timer is 5-60 minutes;
The discreteness of setting the timer time is 5 minutes;
Power supply 220 V, 50 Hz;
Overall dimensions 290x200x155 mm;
Weight 2 kg;
The service life of the device is at least 5 years. The average time to failure is at least 3000 hours;
According to electrical safety, the device is classified according to GOST R 50267.0-92: portable, normal housing, long-term operation, class II type BF for the case of mains power.
Useful videos on our channel!
Awards and achievements
About 80% of Russian medical institutions (more than 18,400) are equipped with magnetic therapy and diagnostic equipment of the company
Medal to them. Academician P. L. Kapitsa for scientific discovery, priority of 1996.
Award of the Government of the Russian Federation in the field of science and technology for 2004
Grand Prix of the International exhibition-Congress "High Technologies. Innovation. Investments" Saint Petersburg.
Grand Prix of the II International Congress of Inventors (Nomination "Invention that gives a push forward") Thailand.
Award of the Government of the Russian Federation in the field of science and technology for 2012
History of the "TPP Center"
In 1991 in I.P.Pavlov Institute of Physiology of the Russian Academy of Sciences on the basis of the Sector of Transcranial Electrical Stimulation Mechanisms, headed by Doctor of Medical Sciences, Professor V.P. Lebedev, the scientific and practical center “Protective Mechanisms of the Brain” was created for the industrial production of TES-therapy devices. For six years, under the license of the authors of the method, the devices were produced under the brand name “ETRANS” at the Yaroslavl radio Plant. In 1996, the scientific and practical center was transformed into the TPP Center LLC, which began assembling the devices «TRANSAIR» (Transcranial electrical stimulator for Analgesia, Immunity and Repair) on its own production base.
Lebedev V. P.
Doctor of Medical Sciences, Professor, Honored Scientist of the Russian Federation, Academician of the Russian Academy of Natural Sciences, laureate of the USSR State Prize, twice laureate of the RF Government Prize in the field of science and technology; author of more than 400 scientific papers and 15 inventions.
Candidate of Technical Sciences, twice laureate of the Russian Government Prize in science and technology; author of 80 scientific papers and more than 20 inventions, the main developer of equipment for thermal power plant therapy.
Kade E. A., Eliseeva L. N. Kuban State Medical University, Department of Faculty Therapy, Krasnodar Introduction. The basis of the pathological process in rheumatoid arthritis